### Tube Temperature Supervision Load Log

This Tube Load Table displays all loads to the tube and housing.

The table disappears after switch off or warm start.

time indicates at which second the event has been logged after turn-on

tube tube number (1...3)

tube disk energy [Ws] parameters

peak load [W] max peak load

rotor energy [Ws] acceleration and brake energy in one package

focus filament used for the exposure

VARIO focus is displayed as large filament at a ratio of 20%, 35% and 50% of small focus, as small with 65% and 80% of small focus