Explanation:

# Warning! Radiation is released during the procedure

The door of the exam. room must be closed.

The collimator of the selected tube must be closed.

The II should be protected against X-ray with a lead apron.

If possible release the exposures from a radiation protected area.

Procedure:

Carry out the adaption for each tube and focus.

Select the Focus you want to adapt.

Select the "START" button to start adaption.

Press the PREP/EXP button or the FOOTswitch to enable X-RAY until the o.k. message appear in the "result" window.

Accept the BoostAdaptation and enable X-RAY until the message in the "result" window appear again.

Proceed the adaptation with the next focus.(if necessary)